Welcome to Smart Home Forum by FIBARO
as you can notice parts of Smart Home Forum by FIBARO is not available for you. You have to register in order to view all content and post in our community. Don't worry! Registration is a simple free process that requires minimal information for you to sign up. Become a part of of Smart Home Forum by FIBARO by creating an account.
As a member you can:
- Start new topics and reply to others
- Follow topics and users to get email updates
- Get your own profile page and make new friends
- Send personal messages
- ... and learn a lot about our system!
Smart Home Forum by FIBARO Team
Jan Hoogendoorn 0
I have a FIBARO Smart Implant device attached to a SenseAir tSENSE CO2-meter which sends the 0-10V voltage signal of the CO2-meter to Domoticz. A virtual sensor "VoltageCO2" contains the voltage from the CO2-meter and a dzVents script converts the voltage to the virtual sensor "CO2 Woonkamer".
There two types of 'sending settings' for the Smart Implant:
The problem with 1) is that everything goes right when changes are rather fast, but when changes are very slow, there seems to be no sending when the critical percentage is exceeded. Method 2) actually does not function at all and I discovered it only after a long time, constantly looking at what was happening. The data are received in Domoticz at the set interval, but in a flash of a second they are zeroed (so when you just look at the values at random intervals, you think that there is no receiving of reading at all).
Now my question is: is this malfunctioning of Domoticz or malfunctioning of the Smart Implant? And if it is the Smart Implant, can it be fixed?
Got more insight in the behavior of the Smart Implant
Link to comment
Share on other sites
0 answers to this question
Join the conversation
You can post now and register later. If you have an account, sign in now to post with your account.